We implement three packs of products, shelf life for one year (from the date of delivery), In the shelf life, the user to comply with the conditions of the use of products and transportation regulations, products such as quality problems, the company is responsible for free repair or replacement.
Do the company's products have inventory?
Most of the products are in stock, but because of the special requirements about radar production from the customer, so the stock is not large.
How to use RS232 serial interface?
Use M62 for the setup of the RS232C serial interface.
The question of measuring error
* extended straight pipe section or the installation of the adjustment device
* check the power supply and timely submission
* check the pipe inner diameter correction instrument factor
* clear or replace the sensor
The sensor sends an abnormal sound
* adjust the flow or replace the large diameter instrument;
* adjust flow and increase liquid pressure;
the flow meter donnot display
* check if the power is connected;
* check the power fuse is in good condition;
* check whether the power supply meets the requirements;
Leave a Message
If you have any suggestions or opinions about our products,please leave a message,and we will immediately answer your questions. Thanks for your support.